Motorized stage connection
Celly motorized stage setting up instruction for Olympus CX-23
- 1.Remove manual stage and focus knob
- 2.Connect the focus module
- 3.Connect XY-stage
- 4.Connect cables
See photos below.
Unscrew 3 screws at the base of the manual stage to remove it.
Unscrew 4 screws at the stage-holder to remove plastic handle.
Manually unscrew the right fine focus knob.
Focus module, 4 screws and a screwdriver. Top rectangle - z-axe connection place, bottom rectangle - microscope backside connection place.
Slide the shroud off and put the motor clutch on the knob screw.
Connect the module to the microscope backside using 4 screws.
Microscope backside after the module connection.
Finally, tighten the screws on the motor clutch and close the shroud.
XY-stage, 3 screws, screwdriver. Rounds: connection area, platform ledge.
Connect the module to the manual stage place using 3 screws.
For Olympus CX-21 model screws are screwed in from the underside of stage
Make sure that the platform ledge falls into the focus module groove.
In some models, the control unit is combined with the focus module, so the z-axis does not need to be connected
Connect signal cables to the control unit matching the red dots on the cable and connector.
Connect the power cable and turn on the control unit.
Almost done! Let's check the CBC test. Set blood smears in the tray (thin part away from you), apply oil to the center of the slides.
Before the app launch move up the stage to 100x lens level
Open the Celly App -> configure analysis -> Type = Blood (CBC).
Check the settings (gear icon):
- number of cells: 100-120
- sensitivity: 65-75%
- crop: 0.15-0.20
- minimap: Off
- Motorized Stage: On (can require app restart)
- Motorized Slide Scan: Off
- Adapter: Other
- Recording: Off
Start your test 🔴 and enjoy automation!